« on: June 06, 2013, 09:40:00 am »
Euge, thanks for the tip. I'll keep my eye out for the SideKIC.
This section allows you to view all posts made by this member. Note that you can only see posts made in areas you currently have access to.
My hydrometer (apparently) lost its calibration somehow. I know I checked it a year or two ago, and it was correct, but I checked it today in 60*F water and got 1.005. Do I just subtract 5 points off any measurement? Why/how would a hydrometer lose its calibration?